ASTM D – Standard Test Method for Dielectric Breakdown Voltage of Insulating Oils of Petroleum Origin. Using VDE Electrodes. ▫ IEC Insulating. Page 1. Page 2. Page 3. Page 4. Find the most up-to-date version of ASTM D at Engineering
|Published (Last):||13 March 2016|
|PDF File Size:||20.44 Mb|
|ePub File Size:||11.48 Mb|
|Price:||Free* [*Free Regsitration Required]|
Avoid touching the electrodes or the inside of the cell after cleaning. Particular reference should be made to the general precaution statement of this test method.
After careful inspection, any electrodes from which pitting cannot be removed by light buffing should be discarded, as more refinishing would destroy the electrode contour and dimensions shown in Fig. If the breakdown voltage is in the expected range for this conditioned oil, the cell is considered properly prepared for asym other samples.
Contact with us now Your mail: It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Request a Free Sstm Right Now. Type text as it appears:: If the test of any sample is below the breakdown value being used by the operator as a minimum satisfactory value, drain the cell and flush the cell with new oil of the type to be tested that is filtered through a 5-micron filter and containing less than 25 x1816 moisture before testing the astk specimen.
Plastic bottles are permeable and moisture content of the sample may change resulting in a measurable difference when compared to samples collected in non-permeable containers.
Clearance from the electrodes to all sides, bottom, cover or baffle, and any part of the stirring device is at least This test method should be used as recommended by professional organization standards such as IEEE C The residue from the buffing should be removed by repeated wiping with lint-free tissue paper saturated with a suitable solvent, followed by solvent rinsing or ultrasonic cleaning.
What are your requirements. It has been found to be especially useful in diagnostic and laboratory investigations of the dielectric breakdown strength of insulating liquid in insulating systems. It is important to avoid touching the electrodes or the cleaned gage with the fingers or with portions of the tissue paper or chamois that have been in contact with the hands.
Pharmaceutical laboratory apparatus stability chambers, tablet disintegration, tablet dissolution, tablet hardness, V-blenders, Measurement of oil temperature: The dielectric breakdown voltage serves to indicate the presence of contaminating agents such as water, dirt, cellulosic fibers, or conducting particles in the liquid, one or more of which may be present in significant concentrations when low breakdown voltages are obtained.
Conduct a voltage breakdown test on a specimen of this oil in the manner specified in this test method. For referee tests or tests that will be used for close comparisons, the laboratories shall agree in advance on the spacing for the tests and ensure that all other requirements of this test method are met.
Calibrate and label automatic controls in terms of rate-of-rise. If a dielectric breakdown does not occur during any of the consecutive breakdown tests using the 2 mm spacing or the sample is not adequate for the 2 mm spacing test cell a 1 more or less 0.
Petroleum Products and Biofuel lab apparatus baths, calorimeters, color measurement, density meters, flash point, hydrogenators, oil testing, penetrometers, surface tension, vapor pressure, viscometers and rheometers, A test cell having a capacity of 0.
A lower than anticipated value is considered as evidence of cell contamination; then repeat the cleaning and the breakdown test with clean dry oil. Right-click to open the PDF in a new window or download it: The spacing agreed upon shall be measured with the gage that corresponds exactly to the selected spacing within tolerance stated above for the gage.
Providing the finest test equipment solutions since 1992
After thorough cleaning, flush the cell with new oil of the type to be tested that is filtered through a 5-micron filter and containing less than 25 ppm moisture. A cell having a capacity of 0. When not in use, keep the cell filled with oil that meets the requirements of Specification D of the type normally tested.
General Use laboratory apparatus balances, baths, blenders, centrifuges, chart recorders, freezers, fume hoods, furnaces, glassware washers, glove boxes, mixers, moisture testing, ovens, pumps, shakers, vacuum equipment, water purification This test method is applicable to insulating asstm commonly c1816 in cables, transformers, oil circuit breakers, and similar apparatus as an insulating and cooling medium. Adjustment and Care of Electrodes and Test Cell 7.
Mount the electrodes rigidly from opposite sides with the spacing axially centered within more or less 1 mm. Alternatively, the cell may be stored empty in a dust-free cabinet.
Email Us or call toll-free: Refer to Terminology D for definitions used in this test method. Determination of the dielectric breakdown voltage of insulating oils provides an early detection method for any reduction in the insulating properties.
The equipment should produce a straight-line voltage-time curve over the operating range of the equipment. Lab Equipment for Sale: After adjustment of the spacing, rinse the cell with a dry hydrocarbon solvent, such as kerosine or solvents of Specification D The voltage in the test circuit should be stable enough to be unaffected by varying current flowing in the capacitive and resistive paths of the s1816 circuit.
Koehler K 75kV Dielectric Strength Tester (ASTM D, D) | Labequip
The impeller, located below the lower edge of the electrodes, rotates in such a direction that the resulting liquid flow is directed downward against the bottom of the test cell. Reinstall the electrodes in the test cup and adjust spacing and clean in accordance with 7. Tightly seal and shield from light until ready to be tested.