Add Attachments to Content Item ‘ASTM D Glass Transition Temperature ( DMA Tg) of Polymer Matrix Composites by Dynamic Mechanical Analysis (DMA) . ASTM D Glass Transition Temperature (DMA Tg) of Polymer ASTM D has become a widely used test standard in many companies due to. ASTM D用DMA测试Tg_能源/化工_工程科技_专业资料。Designation: D – 07?1 Standard Test Method for Glass Transition.
|Published (Last):||8 November 2010|
|PDF File Size:||4.15 Mb|
|ePub File Size:||19.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
Referenced Documents purchase separately The documents listed below e7028 referenced within the subject standard but are not provided as part of the standard. Current edition approved Dec.
ASTM D7028 – 07(2015)
A new or spare specimen shall be prepared from the same material package and tested to replace any specimens rejected per this paragraph. Sampling and Test Specimens 8. Examples are shown in this appendix to provide graphical illustrations of how to select the two tangent lines.
The temperature at which a signi? The values given in parentheses are non-standard mathematical conversions to common units that are provided for information only. Typically the thermocouple should be as close to the sample as possible. Using the approach of this test method the intersection point is drawn as shown in Fig. Summary axtm Test Method 4.
The DMA Tg value is frequently used to indicate the upper use temperature of composite materials, as well as for quality control of composite materials.
Asstm the testing is part of a designed experiment, other sampling techniques may be used if described in the test plan. Any deviations from the standard orientation shall be reported and the test results noted as non-standard.
Before the transition the storage modulus is relatively constant, the sigmoidal change during transition is clear, and after the transition the storage modulus is relatively constant.
In general, for a given material, a higher testing frequency produces a higher DMA Tg value than this standard, while use of the resonance mode will yield a different DMA Tg that may 7d028 either higher or lower than the standard.
A number in parentheses indicates the year of last reapproval. Terminology D de? Opposite surfaces must be essentially parallel and intersecting surfaces perpendicular.
Table 1 aatm the precision statistics generated from this study as de? Any deviations from this heating rate shall be noted in the report and the result shall be reported as non-standard.
Other loading modes such as tensile, torsion or asttm may produce different test results. Should the results be retained, then these conditions shall be described in the test report. These values indicate that the material factors did not have a signi?
The peak temperature of the tangent delta curve Tt is identi? Test Method E speci? Weigh the specimen to the nearest milligram 0.
Weigh the specimen after the test to the nearest milligram 0. Using this approach the intersection point is drawn as shown f7028 Fig. Work Item s – proposed revisions of this standard. Determination and Report of Procedures.
During the glass transition, the storage modulus of the composite material is signi? The span-to-depth ratio, ply orientation, and ply stacking sequence of a specimen with respect to the testing? Record any visual anomalies observed.
D Practice for Plastics: If a non-standard frequency, or the resonance mode, is used, the non-standard approach shall be described in the report and the test result recorded as non-standard. A non-standard DMA Tg de?
The equipment must be calibrated in the same loading mode as will be used for testing, either dual cantilever or three-point bending. Your comments are invited either for revision of this standard or for additional standards and should be d7082 to ASTM International Headquarters. Dynamic Mechanical Proper- ties: It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability d7208 regulatory limitations prior to use.